It is a new kind of opening high frequency jet ventilator controlled by singlechip, it adopts opening ventilation mode with high frequency and low humidifying volume so that the airway pressure is low with little effect on blood stream recycling to heart and it helps to increase the blood flux from the heart and to promote the oxygen combination. The operation does not need to keep the pace with respiration, the device can supply oxygen accessorily and controlled ventilation as well. When ventilating, the suction in airway can be adjusted at the same time.
A. It is widely used for the respiratory management, treatment and first-aid resuscitation of animal such as cattle, horse, tiger, elephant, panda and so on.
B. Multiform switching mode for jetting oxygen through the nose, the device can improve oxygen density rapidly and has an obvious effect on the treatment for all kinds of hypopnea.
C. The device can ventilate through the endotrachea tube and its operation does not need to keep synchronous with respiration. It can be used for tracheotomy in an emergency.
Frequency: 10~150 breaths/min
I/E ratio: 1: 1~1: 3 and inverse ratio
Driving Pressure: 0~0.20MPa
Process ventilation: T=10min,
PRO I: F(150): F(35)=7: 3,
PRO II: F(35): F(150)=7: 3,
Alarm: Alarm with light and sound for no power, and last 6 seconds if no oxygen
CPAP: Not less than 0.05-1.0KPa
PEEP: Not less than 0.05-0.5KPa
Oxygen Concentration Adjustable: 40%-90%
Jetting volume range: 0-4000ML/time
Pressure Regulator Limit: 0.30MPa
Ventilation Mode: HFJV(High Frequency Jet Ventilation); CPAP(Continuous positive airway pressure), PRO I and PRO II ventilation, HFJV+SIGH ventilation; HFJV+PEEP ventilation and so on.