It is a ventilation device for infant with novel structure and good performance, and chooses opening ventilation mode with high frequency and little tidle volume, so the airway pressure is low.
A. The tidal air can letout with constant pressure, constant current or accurately adjustable flow.
B. Opening ventilation, opening intubation makes it not need casing bag, little resistance as expiration is suited for the ventilation management of infant.
C. Evident effectiveness of treating ARDS can be shown with inserting small tube and jetting oxygen through nasal vestibule, which is secure and doesn't cause pressure wound easily.
D. PEEP and humidifier volume can be adjusted accurately.
E. Alarm system with sound and light when no oxygen, poor of electric quantity or low airway positive pressure.
F. Heated humidifier can be used for good effectiveness of humidifying.
G. Accumulator jar, it can be used while being charged and can last for more than four hours.
Frequency: 10~150 breaths/min
I/E ratio: 1: 1, 1: 1.5~ 1: 3 and inverse ratio
Driving Pressure: 0~60KPa
Pressure Atomizer Rate: >=6ml/h
Process Ventilation: T=10min,
ProI: F(150): F(35)=7: 3
ProII: F(35): F(150)=7: 3
Storage Battery: 4Ah, Automatic charge when 12V
Alarm: Alarm when battery voltage is lower than 11V, alarm lasts 6 seconds if no oxygen, Alarm with audible signals and flashing light when the airway pressure is over 0.8KPa
Oxygen Concentration Adjustable: 21%-90%
Jetting volume range: O~600ml/time
Ventilation Mode: HFJV(High Frequency Jet Ventilation); CPAP(Continuous positive airway pressure); HFJV+PEEP(Positive End-Expiratory Pressure), HFJV+SIGH(Sigh), Process ventilation I and Process ventilation II