servo-controlled by micro-computer; quite operation windows can reduce the disturbance to the baby during nursing; inclination of the bassinet can be adjusted, touch switch panel with LED digital display, set-temp, real-temp are displayed separately; multiply self-checking alarms with audible and visible; heating power rate indicates 0-100%; whole aluminum water tank, and it can be filled with water or discharge; blue light radiant box can give the baby jaundice phototherapy treatment when keeping warm on baby with blue light timer; RS 232 connector, data output.
Hood, infant bed, temperature controller, skin temperature sensor, water tank, air filter, blue light radiant box, Guardrail, I.V. Pole, cabinet.
Power supply : AC220V，50HZ
Power input: 600VA
Air Temperature control range：25℃~37℃
Skin Temperature control range：34℃~37℃
Sensor precision: ≤0.3℃
Changing of temperature: ≤0.5℃
Uniformity of temperature: ≤0.8℃
Temperature rising time: ≤30min
Noise level: ≤55dB（A）
Inclination of bassinet: 0°~5°
Average radiation of blue light: ≥500μW/cm2
Timing range of blue light timer：0~9999 hour 59 minutes
Accumulative error of blue light timing：≤0.002%
audible and visible alarm indication, and it can indicate the real temperature and cut off the power supply of heater (except for the low deviation alarm), meanwhile, the skin temperature indication window can indicate the alarm code. For the alarm 1-6, the relevant light is on, and for the alarm 7, all alarming lights are on; all alarm sound can be cancelled except for the power failure alarm and the system alarm.
Over-temp alarm: when the air temperature control ≤38℃, skin temperature control≤40℃；
Upper Deviation alarm: air temperature control +3.0℃, skin temperature control +1.0℃；
Lower Deviation alarm: air temperature control -3.0℃, skin temperature control -1.0℃；
Sensor alarm: when the sensor is disconnected, open circuit, short circuit or put on the wrong place;
Fan alarm: when the fan is blocked, speed ≤1000rpm or stop working;
Power failure alarm: when there is no power or the power cord is disconnected.
System alarm: when there is E2ROM malfunction inside of temperature controller.